Next: MODIFY CABLE
Previous: MODIFY BASELINE
CLIC\MODIFY BUG number
This command is reserved for the correction of nasty bugs in the data
acquisition. The number is used for identification.
Current accepted values:
1 Affecting data taken between 11-nov-1995 and 09-jan-1996. Bug in
RDI found by S.Guilloteau. The continuum data was erroneously cali-
brated in the real-time aquisition. As result, the continuum data
did not agree with the average of the spectral data. For instance
plotting data with "set sub C02 L02" as a function of time gives
different results for C02 and L02. The result would affect line in-
tensities if, as is usually the case, the amplitude calibration is
done with the continuum subbands.
Correction: the current index should contain both atmosphere cali-
brations and correlation data; the data file should be open for both
input and output. This command should not affect data taken before
11-nov-1995 and after 09-jan-1996; be careful for the last day since
the hour is not checked, and RDI was corrected at 3pm.
2 No other bug so far!